(click to download original file)
764 x 941 ( W x H )
Carol Clayton, Leeds
Date of submission
Description of artefact
Grey level fades off at lateral edges, blotchy lighter centre. More obvious on windowing. Visible with all T/F combinations and grid in or out. Not solved by detector recalibration.
Equipment involved and any image enhancement applied
Hologic Dimensions and Selenia.
Cause and/or corrective action
Detector replaced. Cause unknown?
Add new comment
PHP Photo Album