CR IP crack
CR IP crack.jpg
(click to download original file)
1000 x 1000 ( W x H )
Anna Burch, Cardiff
Date of submission
Description of artefact
Uniformity test showing white line at corner of IP,
Equipment involved and any image enhancement applied
Fuji CR 24x30 IP, Fuji Profect reader, Siemens Mammomat 3000. 5x5cm section of original image, windowed.
Cause and/or corrective action
Probably caused by crack in CR plate.
Add new comment
PHP Photo Album